Due to medical law restrictions, this product cannot be sold online, please contact LINE customer service
Roche SARS-CoV-2 Antigen Self Test
SARS-CoV-2 Rapid Antigen Test（Professional Use）
Anti-epidemic project approved to import No. 1106014948
Each box contains a total of 25 antigen detection reagents
Test results can be obtained within 15 to 30 minutes
Can detect alpha, beta, delta variant viruses
Sensitivity: 95.5% (Ct value≤30)
The benefit of having a SARS-CoV-2 antigen test available
The benefits for society, healthcare systems and workers:
The benefits for the general public:
Testing is possible everywhere: Decentralized testing minimizes the possibility of exposure of healthcare workers to potentially infected individuals while protecting healthcare capacity.
Reliable answers: High test quality and performance.
A tool to safely re-open the economy: Tested individuals might be able to travel and participate in events, while reducing the risk of spreading the virus.
Quick results: Faster results compared to laboratory testing.
Real-world evidence: Independent performance studies with large sample sizes, including professional collection, self-collection and self-testing groups.
Easy repetitive testing: No need to visit a healthcare professional or schedule an appointment.
Easy handling procedure: Reduced handling errors and waste.
Increased user protection: Virus inactivation by extraction buffer after 2 minutes.
Peace of mind: Worrying symptoms can be put into perspective.
Testing the quick and easy way
1 Collecting a sample (nasypharyngeal swab)*：Insert a sterile swab into the nostril of the patient and then rotate the swab 3-4 times against the nasopharyngeal surface. Withdraw the swab from the nasal cavity.
2 Preparing a sample: Insert the swab into an extraction buffer tube. While squeezing the buffer tube, stir the swab more than 5 times.
3 Remove the swab while squeezing the sides of the tube to extract the liquid from the swab.
4 Press the nozzle cap tightly onto the tube. Continue with 3a. Performing a test.
5 Place the test device on a flat surface and apply 3 drops of extracted sample in a 90° angle to the specimen well of the test device.
6 Read the test result at 15 to 30 min. Warning: Risk of incorrect results. Do not read the test result after 30 min.
7 A colored line appears in the top section of the result window to show that the test is working properly. This is the control line (C). Even if the control line is faint, the test should be considered to have been performed properly. If no control line is visible the test is invalid.
In case of a positive result, a colored line appears in the lower section of the result window. This is the test line (T). Even if the test line is very faint or not uniform, the test result should be interpreted as a positive result.